Technology that delivers clear, accurate data in simple ways. Intuitive tools that are simple to use. And support you can count on without question. The MAC connects seamlessly with your existing hospital environment, immediately making it easier to acquire, print, store, and transmit ECG data. Your busy practice needs simple solutions. Busy clinical practices need streamlined data connectivity, without the time-consuming complexity. Are you thinking about an upgrade of your ECG equipment?
Resting Ecg. Ecg Machine. Ambulatory Ecg. The device begins to acquire the ECG. Accept the reading and prints the RR Analysis Report on the thermal printer. Press Save to record your settings.
- mac os x backup software.
- living earth hd 1.0.1 cracked from mac app store.
- brooklyn bowl mac n cheese!
Discards the reading and returns to pre-test status. Press Start Test. Press Print. For more information on any of these options. Supported equipment includes the following: Instead of signaling the equipment when a test phase changes. You can also manually override the equipment from the ECG cart keyboard.
Stress tests include the following parameters: Stress Testing 10 The Stress mode is an optional feature that allows you to conduct stress tests with any of the following devices. The system emits a tone to instruct the patient when to take a step. When a test phase changes. Unsupported treadmills Unsupported equipment does not connect to the ECG cart and ergometers system.
Master Step This equipment does not connect to the system. You can control the equipment through this connection. It also offers several configuration options. You can select any of the following report formats: The rate-pressure product is calculated by multiplying the systolic blood pressure with the current heart rate.
The product is then divided by one hundred. Displays in red when the system is in manual mode. Load is displayed in watts. Sweep Speed Changes the speed of the waveform on the display and printout. Changing the measurement also changes the speed of the wiper bar on the display. Item Feature Description Number 6 Phase Name of the current test phase and its total duration in minutes and seconds. It defaults to the setting selected on the Stress Setup window.
In addition. Printer Leads Selects which leads to include in the printout. More Toggles between the first and second row of options. Option Description Select Protocol Selects a predefined set of test criteria. Conducting a Stress Test with a Treadmill or Ergometer Use the following instructions to conduct a stress test with a treadmill or ergometer. For information on the Stress Mode interface. The process is essentially identical for all devices with only minor differences between supported equipment. Target HR Enter the maximum heart rate calculated for the patient based on weight.
More Toggles between the first. Deviations for specific accessories are noted where appropriate. Report Format Selects the components and episodes to include in the report. When conducting stress tests on a supported treadmill. The ECG cart system monitors the heart rate against this target. Conducting Stress Tests There are two basic processes for conducting a stress test: On supported treadmills. With supported equipment. When the exercise phase is complete.
When using an ergometer with remote start. When you are done. On supported ergometers. With unsupported equipment. For more information on setting these options. Continue to record the ECG as needed. Adjust the stress options as necessary: This includes the speed and gain. On unsupported treadmills and ergometers. Continue to monitor the patient and record the ECG until the device stops.
The previous menu options return. For more information on making these adjustments. During the test. Press the Pretest key. When the patient is ready to begin the stress test. Allow the patient to warm up before beginning the exercise phase of the test. This may be a seated.
- mac os x 10.6 snow leopard iso ita?
- GE MAC EKG Operators Manual | Electrocardiography | Physician.
- mac thunderbolt ethernet windows driver?
- MAC - Resting - Diagnostic ECG - Categories.
- tagliare video avi su mac.
- final cut pro mac price;
Record a preliminary ECG. Begin the pretest phase. Have the patient get on the device. Stress Testing 3. When the recovery phase is over. Stress Testing The menu options change. Begin the pretest phase to allow the patient to warm up. Be sure you enter accurate information for Date of Birth. Remove the leadwires from the patient. Instruct the patient to take a step whenever the system beeps. Repeat from step 3 for the next patient.
You are warned that testing another patient discards the results of the current test. The number of steps is determined by these three parameters. The Report Format window opens. You can either print the current test report or change the report formats. This prevents the patient from tripping on the leadwires during the test. Continue to step Select the options you want to include in the report and press Save. Do any of the following. The report prints with the selected format options. This may be seated. Press the Exercise key to begin the test. Reattach the leadwires to the electrodes.
Stress Testing 7. The median report prints at pre-configured intervals during the post exercise stages. ON message is displayed. The duration of the exercise phase is dependent on the selected protocol: When the last post exercise stage is complete. To enable internal storage. You can store resting ECGs automatically or manually: You can only print arrhythmia and stress ECGs. Internal Storage for ECGs. It provides the tools to: Managing Internal Storage To import data via serial port or modem you need to do the following: Press Import.
Continue with step 4. Imported records have a Sent status of Recv and you cannot edit. Press Print Directory. On the Main Menu press File Manager. The system waits while the external device transmits the records. The directory prints on the writer. The File Manager window opens. The modem initializes. Select the records you want to import from the SD card.
Select the appropriate import source from the following options: The function keys change. Use the following instructions to import a record into internal storage: A list of the available ECGs on the card opens. When the correct records are selected. The system waits while the external device transmits records. The File Manager retrieves all the records that match your search criteria.
GE MAC 2000 Operator's Manual
The Enter Search Criteria window opens. Enter your search criteria. Press Search. To clear the search results. To help you locate a record or a group of records. This enters the File Manager into Select mode. Press Select. Imported records have a Sent status of Recv. Managing Internal Storage 3.
Previewing Records Use the following instructions to preview recorded patient data: Edit the information as appropriate. A window opens with the record for you to review. Press Preview. After reviewing the record. Press Edit. The updated information is saved. You cannot edit the patient data for records that were imported to internal storage.
Use the trimpad to select the record you want to edit. After the information is updated. If you only edit demographic information. For instructions on editing patient information. Press Select and use the trimpad to select the record you want to preview. Deleting Records Use the following instructions to delete all records from internal storage: On the Main Menu, press File Manager.
Do one of the following. Press Delete. A window opens and prompts you confirm that you want to delete the selected record s. Printing Records Use the following instructions to print records: The selected records are printed on the writer. Transmitting Records Use the following instructions to transmit records from internal storage to an external device. Before transmitting a record, you must do the following: Use the following procedure to transmit records 1. You cannot transmit records that were imported to internal storage. Press Transmit.
One of two things happens, depending on the number of locations defined in Communications Setup: Select the correct location and press OK. The maximum number of records you can export in XML format is determined by which storage option is enabled: The SD card capacity and manufacturer determine data transfer rates and storage space. This may affect the time required to read or write to the SD card. It may also limit the number of records that you can store on the card. Setting Up Export Options The requirements for setting up export differ depending on the export method: Exporting Records Once the necessary configurations are complete, use the following instructions to export records from internal storage: Select the record s you want to export.
Records that are imported to internal storage cannot be exported from internal storage in Hilltop or XML formats; those records can be exported in PDF format. If you are exporting to an SD card, insert the card into the SD card slot. Make sure the card has sufficient free space for the selected records and that it is not write-protected. If you do not enter the SD card into the SD card slot, you receive the following warning when attempting to export data to the card: SD Card is not present.
Press the appropriate function key: When the export is complete, one of two things happens, depending on the selected format: Press OK to close the summary window. If you want to select additional records to export, return to step 2 or continue to step 6. In the Select Export Destination window, select the appropriate export destination: When exporting to a shared directory, the device logs on to the directory with the user name and password defined on the Communications Setup window.
If either of those values are incorrect, you receive an error message. Correct the user name and password on the Communications Setup window and repeat the export process. Press OK. To access Basic Setup.
You Are Changing Sites
This chapter describes the settings managed by each function and the process followed by each utility. System Configuration 12 System Configuration provides access to functions that allow you to customize the system settings and to utilities to help manage those settings. After making configuration changes. Basic Setup The Basic Setup function allows you to define the following information: You must add physicians in User Setup before they can be picked as default physicians.
Depending on which options were activated. Configuration changes can cause data loss. Field Description Name The name of the institution. Street The street address of the institution. City The city where the institution is located. Defaults on any patient records created on the system. Referring Physician The physician who referred the patient. Technician The technician who conducted the ECG. Location Location ID where the device is located. Cart Unique cart number of the device. When enabled, simulated temporary waveforms are generated in the resting, arrhythmia, RR analysis, or stress ECG functions.
This is useful for demonstration, training, or testing purposes. This setting clears when the system is reset. Field Comment Power up mode Determines which screen is displayed when the system is powered on. Select a color combination that is legible for you. A grid may make reading the ECG easier. The default is on. Anti-Aliasing of ECG Waveforms Determines whether anti-aliasing is applied to waveforms to reduce distortion caused by the video display.
Field Comment Stress Test Identifies the device used to perform the stress test. You must add each user in User Setup. Auto Logoff Determines whether the system automatically logs the user off after a predefined period of inactivity. See also Auto Logoff Time. This is available only if High Security Mode is enabled. Auto Logoff Time 1— min Determines the length of inactivity, in minutes, before the system logs off the user.
You must activate a LAN option to set this option. Time Server Name Identifies the server with which the device synchronizes its time. Contact your server administrator for this information. Last synchronization at Display-only field that identifies when the last synchronization occurred. Last synchronized from Time Server Display-only field that identifies where the last synchronization occurred. Patient Data Input Device Allows the user to select the input methods to download patient demographics.
The available options are: Data Retrieval Allow users to select which data to query: Retrieve Orders from Allow users to select data origin: Cart, Remote, or Cart then Remote. Use the drop-down arrow to select each setting. Speed Changes the speed of rhythm printing and the wiper bar movement across the display. Measurement is in millimeters per millivolt and includes the following options: The default value is 3 leads: The available values depends on which Display Format is selected. Display Lead Group Determines which group of leads is displayed. Restricting frequencies can help eliminate noise in the waveform.
Other values depend on which of the previous two fields are set. Preview before Analysis Determines waveform preview options. For additional information. If enabled. Limit the sum to 0. Pace Enhancement Increases the readability of pacemaker ECG either by augmenting small pace pulses or by truncating large pace pulses. Values include: The values are: Add a marker 1.
This setting is available only if the MI12 option is activated. In all formulas. For more information on activating options. Options are Hz or Hz. Sequence Name Set the display name for a custom lead sequence. You can select the rhythm leads for all four lead sequences. Values are: II -II. If no format is selected. File Manager Sort by Determines the field by which the File Manager sorts records in internal storage.
If the Auto Export for Pharma is enabled. Available only if one of the communications options is activated. Delete After Transmission Determines whether the ECG is deleted from internal storage after it is transmitted to an external device. This is available only if the M or M internal storage option is activated. This is available only if the user has configured and enabled shared directory settings in the Communication Setup. Auto Execute Singe Order Determines whether the single order requested by the user is automatically downloaded and executed.
If the PDFC option is enabled. The following tables list the arrhythmia settings that are unique or differ from resting ECG. The options are: For all other fields. Arrhythmia Setup The Arrhythmia Setup function allows you to define: In Arrhythmia mode. Options Report are: If you chose any of the following in Lead Sequence. In addition to defining the stress ECG settings. System Configuration are unique or differ from the resting or arrhythmia setups. The label for this field changes depending on the filter type selected. The Mode protocol determines the test phases.
You can create custom protocols by selecting the Edit Protocols button. Valid options are: Tabular Summary Determines whether the tabular report format is included in the stress report. Summary Report Options are: Trend Report Determines whether the trend report format is included in the stress report. Summary Report Determines whether the summary report format is included in the stress report. Episodes Printout in Determines how episodes are presented in the stress report. The parameters differ slightly depending on the device.
Each phase can include multiple stages that define the parameters of the test. Select the stage that precedes the new stage. The Add Protocol window opens. Press Page Down. Edit the duplicate stage as appropriate. Press Add. System Configuration Stress Test Parameters cont'd. Perform step 4 through step The templates are based on the existing protocols. A list of templates opens. An Empty Protocol is also available. Use the following instructions to create a custom protocol: Type a name for the new protocol and press OK.
The Protocol window opens with all the stages from the template. Press Add Stage. You can now add. You can modify the pre-defined protocols to create custom protocols. Perform step 8 through step The selected stage is duplicated. The second page opens. See step 9. For treadmills and ergometers. To add a stage. Select the template on which you want to base the new protocol. Select Edit Protocols and press either Enter or the trimpad. This saves your changes and returns you to the previous window.
To edit a stage. To rename the protocol. Select the stage you want to delete. Press Edit Name. Repeat steps 8 through 10 as necessary. See step Change the name as appropriate. This option is not available when editing a Master Step protocol. Communication Setup The Communication Setup function allows you to define the following settings: The Edit Stage window opens.
Select the stage to edit. The Edit Name window opens. When you are done with the stages. To delete a stage. When the protocol is done. The selected stage is deleted. System Configuration 9. Modify the stage parameters as appropriate.
Об этом товаре
Press Delete Stage. The Edit Stage window closes. Reset treadmills and ergometers to factory defaults at the protocol level. To remove custom protocols. Refer to the table preceding these instructions for a description of each parameter. The protocol is saved and you return to the Select Protocol window. The locations are defined on Page 2 of this Communication Setup Fields table. If this field is set. To access the Communication Setup. This field is displayed only if the M or M Internal storage option is activated.
GE Healthcare - Mac Community, Manuals and Specifications | MedWrench
The following tables describe the settings on Communication Setup. If this field is not set. Share Name Identifies the name of the shared network drive. If this field is checked. The user must be set up on the domain with the appropriate permissions to access the shared directory. Password Identifies the password that the system uses to log on to the shared directory. IP addresses are not supported. This field allows a maximum of 30 characters. Username Identifies the user name that the system uses to log on to the shared directory.
This field allows a maximum of characters. Fields Description Serial Baud Rate Determines the speed at which data is transmitted across the serial communications port when using a modem. Communication Setup—Data Communication Locations Field Description Location Identifies the name of a communication location that receives the transmission from the system.
Fields Description Confirm Re-enter the password in this field to confirm that the password was entered correctly. You can define up to four locations. Device Identifies the type of device to use to transmit data to the location. This field becomes active only after a corresponding location is entered. This system can use several protocols to communicate test data and retrieve patients or orders. It supports retrieving patient demographics and orders and sending patient tests.